SAN DIEGO, Sept. 30, 2013 /PRNewswire/ -- CareFusion issued the following update regarding its previously announced voluntary recall of AVEA ® ventilators.
The company initiated the voluntary recall on Sept. 3, 2013. The FDA has classified this action as a Class I recall. A Class I recall is defined as a reasonable probability of serious adverse health consequences or death associated with use of the defective units.
CareFusion announced a voluntary recall of AVEA ® ventilators regarding barometric pressure sensor compensation when using the neonatal wye hot wire flow sensor. AVEA ventilators may experience an underreporting of tidal volume if used in conjunction with the neonatal hotwire flow sensor. The patient may receive a higher than expected tidal volume. The error is due to lack of barometric pressure sensor compensation.
While all AVEA ventilators fall under this voluntary recall, hospitals in locations at high elevations (higher than 5,000 feet above sea level) that are using the AVEA ventilator in the neonatal patient care setting are at greatest risk of this error. The level of underreporting between displayed flow and actual flow increases proportionately with altitude. CareFusion has received five complaints from a facility that was operating in the neonatal patient care settings and was located higher than 5,000 feet above sea level. No injuries have been directly attributed to this issue.The Food and Drug Administration (FDA) has been notified of this recall. Instructions to Customers CareFusion does not require the return of the devices. CareFusion advises its customers in locations higher than 5,000 feet above sea level using the AVEA ventilator in neonatal patient care settings with a wye hot wire flow sensor to take one of the following actions:
- If using the AVEA Comprehensive ventilator, continue the use of the ventilator using the VarFlex ™ single-patient use sensor (CareFusion part number 50000-40038), and discontinue use of the neonatal hotwire flow sensor until the software correction has been installed.
- If using the AVEA Standard ventilator, discontinue use in neonatal patient care settings and contact CareFusion customer support.
- If a hospital is located below 5,000 feet above sea level no action is necessary at this time.
- Online at http://www.fda.gov/Safety/MedWatch/HowtoReport/default.htm (form available to fax or mail), or
- Call FDA at 1-800-FDA-1088.
Select the service that is right for you!COMPARE ALL SERVICES
- $2.5+ million portfolio
- Large-cap and dividend focus
- Intraday trade alerts from Cramer
- Weekly roundups
Access the tool that DOMINATES the Russell 2000 and the S&P 500.
- Buy, hold, or sell recommendations for over 4,300 stocks
- Unlimited research reports on your favorite stocks
- A custom stock screener
- Upgrade/downgrade alerts
- Diversified model portfolio of dividend stocks
- Alerts when market news affect the portfolio
- Bi-weekly updates with exact steps to take - BUY, HOLD, SELL
- Real Money + Doug Kass Plus 15 more Wall Street Pros
- Intraday commentary & news
- Ultra-actionable trading ideas
- 100+ monthly options trading ideas
- Actionable options commentary & news
- Real-time trading community
- Options TV